There are currently no product reviews.
Perfect quality, highly appreciated service !!!
Fast delivery of the download info, no issues at all. Recommended E-manual shop !!!
hi owners manual- crew,
i am very satisfied about the trade with you.
if i need some other manuals, i will contact you again.
Full Panasonic service Manual, as described, no problems
This place is amazing. Got our manual in a few hours. Print quality is excellent. Even the manufacturer didn't have this manual. The price was excellent. How many more stars can I give them? More than satisfied.
This was a very hard to find manual. The unit is long discontinued and Pioneer doesn't retain every manual for every model. Thanks owner's-manual.com!
5-5. TEMPERATURE COMPENSATION OFFSET ADJUTMENT
Save the temperature data at that time in the non-volatile memory as 25 ËC reference data. Note : 1. Usually, do not perform this adjustment. 2. Perform this adjustment in an ambient temperature of 22 ËC to 28 ËC. Perform it immediately after the power is turned on when the internal temperature of the unit is the same as the ambient temperature of 22 ËC to 28 ËC. 3. When D101 has been replaced, perform this adjustment after the temperature of this part has become the ambient temperature. Adjusting Method : 1. Rotate the AMS knob and display âTEMP ADJUSTâ. 2. Press the YES button and select the âTEMP ADJUSTâ mode. 3. âTEMP = â and the current temperature data will be displayed. 4. To save the data, press the YES button. When not saving the data, press the NO button. 5. When the YES button is pressed, âTEMP = SAVEâ will be displayed and turned back to âTEMP ADJUSTâ display then. When the NO button is pressed, âTEMP ADJUSTâ will be displayed immediatelly. Specified Value : The âTEMP = â should be within âE0 - EFâ, âF0 - FFâ, â00 0Fâ, â10 - 1Fâ and â20 - 2Fâ.
5-6. LASER PPOWER ADJUSTMENT
Laser power meter
Optical pick-up objective lens Digital volt meter
BD board CN110 pin 5 (I+3V) CN110 pin 4 (IOP)
Adjusting Method : 1. Set the laser power meter on the objective lens of the optical pick-up. (When it cannot be set properly, press the 0 button or ) button to move the optical pick-up.) Connect the digital volt meter to CN110 pin 5 (I+3V) and CN110 pin 4 (IOP). 2. Rotate the AMS knob and display âLDPWR ADJUSTâ. (Laser power : For adjustment) 3. Press the YES button once and display âLD 0.9 mW $ â. 4. Rotate the AMS knob so that the reading of the laser power meter becomes 0.86 to 0.92 mW. Press the YES button after setting the range knob of the laser power meter, and save the adjustment results. (âLD SAVE $ â will be displayed for a moment.) 5. Then âLD 7.0 mW $ â will be displayed. 6. Rotate the AMS knob so that the reading of the laser power meter becomes 6.9 to 7.1 mW, press the YES button and save it. Note : Do not perform the emission with 7.0 mW more than 15 seconds continuously. 7. Then, rotate the AMS knob and display âLDPWR CHECKâ. 8. Press the YES button once and display âLD 0.9 mW $ â. Check that the reading of the laser power meter become 0.85 to 0.91 mW. 9. Press the YES button once more and display âLD 7.0 mW $ â. Check that the reading the laser power meter and digital volt meter satisfy the specified value. Specified Value : Laser power meter reading : 7.0 Â± 0.1 mW Digital voltmeter reading : Optical pick-up displayed value Â± 10% (Optical pick-up label)
Iop = 82.5 mA in this case Iop (mA) = Digital voltmeter reading (mV)/1 (â¦)
10. Press the NO button and display âLDPWR CHECKâ and stop the laser emission. (The NO button is effective at all times to stop the laser emission.)
â 32 â
Complete service manual in digital format (PDF File). Service manuals usually contains circuit diagr…