There are currently no product reviews.
I'm very happy that you all are performing an incredible good job. Furthermore what you did is very useful for all people as me that have electronics as an hobby.Thank you!!!!!
Manual found fast and good quality, very helpfull service
8-17-12 Been using the sight for about 6 months. Fast Downloads and top quality
Everything was great, the manual, the response time, the simplicity of the order, and the
Price. The only thing that I could possible say on a negative note would be that the manual I ordered was more for a service tech. There were a lot of schematic diagrams that didn't help me solve the problem. However I would order again and recommend the web sight to others.
I'd been looking for this manual for awhile. Exactly what I needed - and at an excellant price. Thanks!
3. Mode: Playback
test tape P-4-A100 (10 kHz, â10 dB) L-CH MAIN board CN510 R-CH pin 3 waveform of oscilloscope
CD SECTION [TEST DISC LIST]
pin 1 L R pin 2 V H oscilloscope
Use the following test disc on test mode. â¢ CD: YEDS-18 (PART No. 3-702-101-01) or PATD-012 (PART No. 4-225-203-01) Note: 1. CD Block is basically designed to operate without adjustment. Therefore, check each item in order given. 2. Use YEDS-18 (3-702-101-01) unless otherwise indicated. 3. Use an oscilloscope with more than 10MW impedance. 4. Clean the object lens by an applicator with neutral detergent when the signal level is low than specified value with the following checks.
in phase 45Â° good
90Â° 135Â° 180Â° wrong
4. After the adjustments, apply suitable locking compound to the pats adjusted.
BD91 board oscilloscope
Adjustment Location: Playback Head (Deck A). Record/Playback/Erase Head (Deck B).
Procedure : 1. Connect an oscilloscope to TP106 (FE) and TP124 (VC). 2. Turn the power on. 3. Load a disc (YEDS-18) and actuate the focus search. (In consequence of open and close the disc tray, actuate the focus search) 4. Confirm that the oscilloscope waveform (S-curve) is symmetrical between A and B. And confirm peak to peak level within 3 Â± 0.5 Vp-p.
S-curve waveform symmetry
within 3 Â±0.5Vp-p B
Note: â¢ Try to measure several times to make sure than the ratio of A : B or B : A is more than 10 : 7. â¢ Take sweep time as long as possible and light up the brightness to obtain best waveform.